Intelligent pressure transmitters UIB6 can be used in aviation, aerospace, petroleum, chemical, mining, machinery, dams, geology, hydrology, food, medicine, health, water treatment, power, marine engineering and other industries for measurement of liquid, gas, steam,level pressure and differential pressure. Therefore, the selection of the installation location is very particular. When measuring gas, how should the installation location be selected?
1. The installation position of the digital pressure transmitter UIB5 should be well lit and easy to maintain. It is not suitable to install in the location with vibration, humidity, high temperature, corrosion and strong magnetic field interference.
2. The pressure point should be in the upper half of the process pipeline; if it is to measure the vapor pressure, the pressure point should be taken at the upper half of the process pipeline and the lower half of the process pipeline at an angle of 0-45 degree with the center line.
3. The installation position of the pressure source component should be selected in the pipe section where the process medium flow is stable; when the pressure source component and the temperature source component are on the same pipeline, the pressure source component should be installed on the upstream side of the temperature source component ; The end of the pressure source component should not exceed the inner wall of the process equipment and process pipeline.
For details, please contact Qi Huang